Shore Durometer Test Stand
You May Also Be Interested In:
shore durometer test stand
SLX-D Durometer Test Stand, hardness tester:
The test stand construction includes an operating handle, adjusted glass stage, die poise, clip beam of durometer, and carrick column. It is more accurately to test hardness of rubber or plastic by using this product.
1) Fix durometer on the test stand, drive handle
2) Under weight of die poise, durometer the indenter foot on bottom of press head, touch the surface of glass stage completely, the value of durometer is 100+/-0.5 HA. If the value is not 100+/-0.5 HA, please adjust two vernier knobs until the value is 100+/-0.5HA. then place the test sample on glass stage; drive handle
3) Under the weight of die poise, durometer the indenter foot on bottom of press, head touch with the test sample completely. At this time, the value is showed on durometer is the shore hardness of the test sample
4) when you adjust the parallelism, please do not press the tip on glass stage directly. That will damage glass stage.
Please place a piece of even glass board on the glass stage before you adjust the parallelism