You May Also Be Interested In:
YD-2335 computerized microtomeis a new high precision microtome after YD-335 computerized microtome . the machine laid more stress on constructional advantage in machanical-electrical integration, which make machine and electricity perform seperately but non interfering . Thus, they reach section's requirements of high precision and high stability. The machine is an ideal product for morden histopathology section because it's sophisticated and stable machanical construction and exellent drive performance of electric stepping.
1) it adopted the advanced import actuation system so that can slice the tissues more precisely , more reliable and quiet.
2) the LCD display show the slice and trimming thinkness , section counter in chinese and english vertion.
3) it have special fuction of sample rebounding . it can avoid the damage due to the attition between sample and the back of knife so that the slice is more smooth and the life of the knife is more longer .
4) the handwell can be locked in any position to make the section as safe and convient as possible .
5) the angle between sample and knife can be adjusted freely.
6) the waste bath can be disassembled easily
2) Slice thickness range: 0- 60um
3) Thickness for repairing the tissue slice: 0 - 60um
4) Horizontal displacement of the sample: 50mm
5) Vertical dislocation of the sample: 93mm
6) Sample rebounding: 15um
7) Lowest slice adjusting graduation: 1um
8) Tissue slice adjusting graduation: 1um
9) Slip precision: 1 10%
10) Maximal slice section: 50 x 40mm
11) The angle between cyroplate and cyroscalpel is 450wcih made the tissue attatched to the slice easily.
12) maximum diffirence in temperature of freezing stage >=60:C
13) maximum diffirence in temperature of cooling knife >=50:C
14) automatica recovery of refrigeration working state after defrost
15) refigeration race can reach slice working temperature in 4-15minutes